Posted: 2013-03-07, 06:01 Repair the dental handpiece
Today we will discuss a bit about repairing the dental handpiece. As a dentist, it sometimes happens that a new tomb bearing defective shortly after the repair of the handpiece or the speed is not enough. But we want to know why.
I summarize the probable factors below:
the poor quality of the bearing replaced by the dealer;
damage to the handpiece itself (dropped by ememple);
aging some small parts (seal cartridge, NSK, etc.).
the wrong type of bearing incompatible with the handpiece;
the usual problem of handling (do not pay attention to the lubrication, the aging of the peak and the fall of the handpiece);
the problem of water and air (the source of clean water and air);
All the above factors will give a great influence on the life cycle of your handpiece.
So, I propose some measures to avoid as much damage to your dental handpiece.
Replace tip regularly to prevent rotation vacuum;
Prevent the fall of the handpiece;
Consult the owner's manual carefully before use to determine the operating air pressure to clean the handpiece;
Frequently lubricate and clean your dental handpiece.
If you can do what is proposed, the life of your dental handpiece will be extended by several years.
You cannot post new topics in this forum You cannot reply to topics in this forum You cannot edit your posts in this forum You cannot delete your posts in this forum You cannot vote in polls in this forum You cannot attach files in this forum You can download files in this forum